Design Outputs – Truly Control Your Design
By Mark Foley – Engineering Manager
Understanding Design Outputs for Medical Device Developers
If you are involved in the development of medical devices, it’s likely you’ve heard the words ‘design outputs’. Well – What are design outputs? What purpose do they serve? Why should I be concerned about them? These are all just a few questions you will probably ask yourself.
I once asked myself the very same questions. After spending the better part of the last seventeen years in medical device product development, I hope that I can help explain what is often a widely misunderstood part of the process.
Understanding the Design Control Process
I’m sure it’s no surprise that referring to the FDA-regulated process for medical device development known as design controls. The design controls process provides a product development workflow and a set of criteria for which the FDA expects medical devices under development to conform to if a company hopes to receive approval to market their device in the U.S.
What Are Design Outputs?
Design outputs are the work produced at the end of design efforts, such as drawings, material specifications, or manufacturing instructions. They are what control your design. They must also meet design input criteria defined in the planning stage of your project during design verification.
Considering Design Outputs Before Design Verification
I often see little time spent thinking about how design outputs will be verified until after the close of the design effort and the start of the design verification stage. By this time, it’s too late; it’s something that needs to be considered right from the start of design inputs at the beginning of your work.
You will end up wasting time making updates to outputs that have already been released but don’t provide the information you need to verify their inputs. You will also find yourself having to navigate the change control process in a hurry as you try to keep the project on schedule.
I’ve seen projects delayed for months because of things like this. This scenario can be avoided by paying a little more attention to the design outputs and ensuring they contain the information to make verification possible.
Create Accountability for Evaluating Design Outputs and Inputs
I recommend that during your design stage you create a mechanism to serve as a deliverable that will force you to evaluate each design output against each design input before its release. This is a paper exercise that can often be accomplished in a single document. I like to refer to the document as a technical review.
In this document, capture your input, and the acceptance criteria that you will use in verification, checking that the information or specification is contained in your design output. Time spent performing this exercise before closing your design stage and freezing your design will serve you well.
Changes made after a design freeze can often mean that other activities or assessments will need to happen and these activities are sure to drain time from your schedule and money from your budget, which doesn’t do anyone any good.
Partnering With An Industrial Design Company
Partnering with an industrial design company, all phases in the product development plan can be streamlined to accelerate time to market, reduce waste, and meet regulations.
At Goddard Inc., our team of engineers and designers specialize in medical device development, life sciences and diagnostics, and high-tech consumer and industrial technologies. From product conceptualization to production, our team delivers a competitive difference with an on-site partnership model, commitment to quality, unique collaborative expertise, and in-depth employee development programs.
We would love to hear about what you’re working on.
Tell us about your project and let’s begin a conversation about creating products with technical precision and elegant design.